The Maselli beverage analyzers must be operated either with the basic MP01 receiver or the more sophisticated MP02 one. The MP02 receiver is PC Windows based color touch screen panel which communicates with the sensors by means of a RS485 digital connection. The intuitive software gives the opportunity to remotely operate the sensors but as well to visualize all the collected data in a graphic way showing trends and alarms conditions. The MP02 has got all the advantages and versatility of a PC including a complete range of outputs such USB, Ethernet or HDMI for remote display. The unit may be connected to Maselli M8 data collection software in the laboratory.
|Application:||Display and control of qualitative parameters on process lines producing sugar-sweetened or diet carbonated soft drinks, alcoholic beverages, non-carbonated soft drinks, beer and mineral waters|
|Type of function:||Interfacing with IB07, IB08 or Ur29, UR27, UC07, UC08, US01 single Analysis Units. Options for the display, calibration, diagnostics and setup of operating parameters.|
Acquisition, via Analog inputs, of mV, mA, Pressure and temperature signals.
Activation of two programmable alarm contacts.
Acquisition of two programmable input contacts.
Management of 3 serial outputs one of which is personalizable via HMS module.
|Power supplies||Electric: |
DC 24V 10% 7A Max
Variable absorption depending on the appliances connected.
Connection box without Transformer (optional):
Power supply according to MP02 specifications.
Terminal board connection.
Connection box with Transformer (optional):
AC 100…240V 10% 50...60Hz 170VA
Terminal board connection.
2 active output channels 0...20mA or 4...20mA (470 max.) configurable in the "Max. and Min. full scale" values.
4 auxiliary input channels.
RS485 for connection to Maselli analyzers
PROFIBUS DP or ETHERNET/IP (optional)
2 configurable inputs
2 relay outputs for alarm signals with contacts of a maximum capacity of 1A/24V DC/AC
For any software updates or recipe downloads required
RJ-45 for external connection (laboratory installed M8)
|Execution:||Chassis in AISI 304 stainless steel with removable front panel, wall or post mounting system, 10” touch screen monitor|
|Electronic section:||“CPU” with PENTIUM® INTEL® microprocessor (or similar, depending on technological evolution), 4 GB RAM memory, Video Card, Modem, 3-button mouse.|
10” monitor with analog resistive touch screen, resolution: 2048x2048
USB V 2.0
IEEE 802.1p and 802.1q supported
10/100/1000 IEEE 802.3 compliant
|Dimensions and weight:||340 (w) x 300 (h) x 160 (d), 5.5 kg|
|Accessories:||Different fasteners for installation on the wall or piping (diameter 40 - 100 mm).|